FTM-50A laser displacement sensor (industrial economy)
The FTM-50A laser displacement sensor has a maximum distance of 50m and an accuracy of ± 3mm.
FTM-50A laser ranging sensor has a wealth of industrial data interfaces (RS232, 485, 422, 0-10V, 0-5V, 4-20mA, etc.), strong anti-interference ability, widely used in the steel industry, metallurgy , Automotive industry, printing industry, food industry and other industrial control and various field monitoring and testing sites.
1. Reflector measures the distance of various objects;
2. The filter can measure the object under high temperature;
3. Explosion-proof box can be used in explosion-proof environment;
4. The swing motor can be used for 2D contour measurement.
Laser characteristics: red laser diode
Laser level: Level 2
Spot type: dot / line
Working voltage 5V
3. Mechanical parameters
Housing material: aluminum alloy coating or plastic spraying
Window: glass window
Wiring method: aviation plug or reserved 2m shielded wire
Length, width, height: 90 × 50 × 25mm
Protection level: IP65
Storage temperature: -30 ℃ -80 ℃
Output mode: RS232, RS485, 4-20mA, 0-10V or TTL;
Input voltage: 5V or 12V;
Working mode: power on or command control.
Fourth, technical indicators
1. Measuring range: 50m
2. Accuracy: ± 3mm
3. Sampling frequency: 2HZ-8HZ
4. Working temperature: -10 ℃ -50 ℃
5. Working voltage: 5V / 12V / 18-24V
6. Power consumption: 1W
7. Light spot diameter: 10mm @ 30m
8. Size: 25 × 50 × 90m
Five, maintenance and upkeep
1. Wash and keep dry.
2. Blow off the dust on the mirror surface.
3. Do not touch the lens with your fingers.
4. Wipe only with a clean, soft cloth. If necessary, wipe it with pure alcohol or water-wet cloth.
5. If the instrument is accidentally wet, do not expose it to the sun, it should be air-dried.
6. If not used for a long time, please turn off the power to avoid affecting the life.
Six, matters needing attention:
1. Do not remove or damage the warning label on the instrument when using the instrument.
2. Avoid direct laser radiation to the eyes, which will cause instant blindness in the eyes.
请勿在小孩周围操作仪器或让小孩自行操作，避免伤害眼睛。 3. Do not operate the instrument around children or allow children to operate by themselves to avoid eye damage.
4. Do not try to change the performance of this instrument, which may cause serious laser radiation injury.
不要维修或拆解本仪器，非专业人员维修可能导致严重的激光辐射。 5. Do not repair or disassemble the instrument. Non-professional maintenance may cause serious laser radiation.